In this episode Shahriar repairs an Anritsu MS9710B Optical Spectrum Analyzer. The instrument intermittently does not boot and when it does it generates a Grating Mirror error. Some investigation reveals that the problems may simply be poor internal cabling connections which can be resolved by cleaning all connectors and re-building the unit. The LCD screen’s CCFL have also failed. An LED back-light replacement provides the screen with a vibrant and bright colors.
The unit is very compact (which is unusual for an Optical Spectrum Analyzer) and makes it ideal for bench-top working environments. The broad-spectrum built-in reference laser can be used to calibrate and align the instrument which proves to work perfectly. After calibration, a laser diode from an Agilent Lightwave Transmitter is measured. Interestingly it can be seen that the unit produces two tones at ~1nm apart which indicates the laser operates in two modes. A very interesting result given that the wavelength meter only reports one tone at the average wavelength of the two modes.
The Signal Path